by means of the arrow keys, select "Access w-out pass" (access without password - user level), confirm by pressing ENTER.
The other levels require password reserved to the Technical Service, to the Manifacurer, etc.
The menu accessed without password is the following:
Choose "LoadController" and press ENTER: the following menu is shown:
Choose "ControllerParam" and press ENTER: the following menu is displayed:
Choose "SetPointW1" and press ENTER:
Curr: it shows the current set-point; use the arrows keys to change it.
NOTE: the available range depends on the probe used; the measure unit of the detected data and its relevant limits are set by "Service"
Once the new set-point is set, confirm by pressing ENTER, otherwise exit without changings by pressing ESC.
Press ESC to exit the set-point programming mode.
Once the temperature set-point W1 is imposed, set the Switch-on (SDon) and the Switch-off (SDoff) point of the 2-position controller:
To set these values, select the item SD_ModOn (SDOn), by scrolling down the "Load controller" menu with the arrow keys and press
Access w-out PW