4 Configuration Settings Appendix
Configuring Predictive Temp
pToral, pTaxil, pTrect Settings
Measurement Settings: Main Setup -> Measurements -> <pTemp Label>
pToral (or other label)
Predictive Temp Configuration Implications
Label Lets you select the label you want to configure. This is not a setting and cannot be preconfigured.
Value Lifetime Lets you configure the life time of a predicted temp value. After the configured life
time, a predictively measured Temp value disappears from the monitor screen, as well as from the screen of
a connected Information Center. This setting applies to all predictive temp labels.
Mode Lets you switch between Continuous and Predictive measurement mode. This switch
only applies when monitoring, it is not a setting and cannot be preconfigured. See the IntelliVue
Instructions for Use for more detail.
Prompt Tones Set this to On if you want the monitor to indicate when it is ready for a new
measurement and when a measurement is finished by a prompt message and a prompt tone.
Unit Lets you select the temperature unit.
Preferred Label Lets you configure which predictive temp label the monitor will use when one of
the following circumstances occurs:
• an oral or axillary probe is connected after a rectal probe has been used before,
• the patient is discharged,
• a coldstart is performed.
<PTemp Label> The On/Off state of the predictive temp measurement cannot be preconfigured. It
is automatically switched On when a predictive temp sensor is connected to the monitor.
MP5, MP5T, MP5SC
C M Profile Adult
x not applicable, the 'Label' is not a setting, see "Predictive Temp Configuration
x 1 h
x not applicable, 'the Mode' is not a setting, see "Predictive Temp Configuration
x not applicable, the Predictive Temp On/Off state is not a setting, see "Predictive Temp