Temp, Trect, Tcore, Tskin, Tesoph, Tnaso, Tart, Tven, Tvesic, Ttymp,
Tcereb, Tamb, T1, T2, T3, T4 Settings
Measurement Settings: Main Setup -> Measurements -> <Temp Label>
Temp (or other label)) x
See "Configuring C.O. (Cardiac Output)" on page 65 for Tblood settings. Tinj has no settings. Taway is
sourced from a VueLink or IntelliBridge plug-in module, therefore no settings can be changed.
Temp Configuration Implications
<Temp Label> The On/Off state of the Temp measurement cannot be preconfigured. The Temp
measurement is automatically switched On when a Temp probe is connected to a Temp socket on the
Unit Lets you select the temperature unit for the temperature label.
Range This setting defines the measurement range for the temperature label. Available choices are
1...45, 11...45, and 35...43.
The configured Range is used for the horizon screen trend application. When a temperature is viewed as
• the high end of the Range defines the maximum temperature that can be selected in the related Screen
Trend menu under Set High Horizon,
• the low end of the Range defines the minimum temperature that can be selected in the related Screen
Trend menu under Set Low Horizon.
MP5 - MP90 (H10/20/40)
C M Profile
x not applicable, the Temp On/Off state is not a setting, see "Temp Configuration
Implications" on page 83.
4 Configuration Settings Appendix
MP5 - MP90 (H30)