PPV Configuration Implications
PPV / Measurement If you set Measurement to Enabled, the user can switch the PPV
measurement On in monitoring mode, provided the set arterial source is available. If Enabled, the
derived PPV may generate a label conflict with the PPV calculated by the CCO measurement, see "CCO
Configuration Implications" on page 67.
If set to Disabled, the PPV measurement is permanently disabled, which means that in monitoring
mode it will not be possible to switch PPV On.
Arterial Source This setting is only available when Measurement is Enabled. Possible
sources are: P, ABP, ART, AO, FAP, BAP. The formula used to calculate PPV is:
Measurement Settings: Main Setup -> Measurements -> C.O.
Tblood High Limit
Tblood Low Limit
1.Setting only available if Transpulmonary method is selected.
2.Setting not available in the U.S.A or in clinical environments under FDA control.
MP20 - MP90 (H10/20/40)
x not applicable, this is not a setting, see "C.O. Configuration Implications"
x not applicable, the C.O. On/Off state is not a setting, see "C.O. Configuration
4 Configuration Settings Appendix
MP20 - MP90 (H30)