4 Configuration Settings Appendix
NBP Alarm Suppr. This setting has an effect only when using SpO
be generated when you measure NBP on the same limb as SpO
to On, the monitor automatically remembers the SpO
INOPs while the cuff is inflated.
Extd. Auto On/Off The default for this setting is Disabled. If Enabled, the SpO
measurement will be automatically switched Off when the SpO
the Global Setting Sensor Disconnct is set to Auto Off (see "Sensor Disconnct" on page 172),
and at least one of the following criteria is met:
• the resulting SpO2 Sensor Off INOP is silenced,
• alarms are paused or switched off. This also applies if alarms are remotely suspended from a connected
• both the SpO
If the sensor is reapplied, the SpO
Note that the setting Extd. Auto OnOff has no effect if a telemetry device or a CL SpO
connected to the monitor via a short range radio connection.
Never set Extd. Auto On/Off to Enabled if continuous SpO
measurement may be inadvertently turned off, when
• the SpO
• all alarms are off when the sensor comes off the finger, or
• the Pulse (SpO
• the user switches all alarms off while the sensor is off the finger, or
• the user switches off the Pulse (SpO
Consider the implications when you configure Extd. Auto On/Off differently for different
C A U T I O N
Measurement Settings Blocks and Profiles.
. Set NBP Alarm Suppr. to On to suppress INOPs that would otherwise
alarm and the Pulse(SpO
measurement is automatically turned On.
sensor is off the patient's finger and the user silences other alarms, or
) and the SpO
alarms are switched off when the sensor comes off the finger, or
. If NBP Alarm Suppr. is configured
value measured before cuff inflation and suppresses
) alarms are switched off.
) and the SpO
alarms, while the sensor is off the finger.
sensor comes off the patient's finger, and
monitoring is intended, because