4 Configuration Settings Appendix
Select the Delete key. This removes this lead from the list of chosen leads.
The order in which ST leads are listed in the Setup ST Leads menu determines the order in which ST leads
are displayed on the monitor screen.
To change the order in which ST leads are displayed,
In the Setup ST Leads menu, choose a lead from the list.
Select the Sort Up or Sort Down key to move the lead up or down in the list.
Changing ST Alarm Limits
The monitor can detect alarms on each ST lead separately, so you can set high and low ST alarm limits
individually for each ST lead. You can also set separate alarm limits for single-lead and multi-lead ST
Configuring QT Analysis
Measurement Settings: Main Setup -> Measurements -> QT Analysis
QTc High Limit
ΔQTc High Limit
QTc High Alarm
ΔQTc High Alarm
QT Analysis Configuration Implications
QT Lead For QT Monitoring you can select one of the following QT Lead modes:
• All: all available leads (I, II, III, V, MCL, V1 - V6) are used to produce a global QT measurement. For
EASI lead placement, directly acquired AI, AS and ES leads are used.
• Primary: the primary lead will be used for QT measurement. If the original primary lead becomes
unavailable or is changed, QT measurement will continue with the new primary lead.
• I,II,III,MCL,V,V1-V6: a single lead selected from all available leads (except the augmented
leads) will be used for QT measurement. QT measurement will stop if the selected lead becomes
QT Analysis This setting lets you switch QT Analysis On or Off.
MP2 - MP90
C M Profile Adult