ST-Index This setting lets you switch the ST-Index numeric On or Off for display. The ST index
numeric (STindx) is the sum of the absolute values for the ST leads V2, V5, aVF. Because it is based on
absolute values, it is always a positive number. If you haven't selected one of the leads V2, V5, and aVF for
ST analysis, the STindx numeric will display a question mark "?".
ST Uses If ST Uses is set to J+60 or J+80, the position of the ST Point is set relative to the
J Point. Change the ST Point by positioning the J Point up to 380ms after the peak of the R-wave.
If ST Uses is set to ST Point, the ST Point can be set directly and independently of the J Point
position. The ST Point can be positioned up to 460ms after the peak of the R-wave.
Note that switching between the settings does not move the J Point position.
Lead I, II, III, aVR, aVL, aVF, V, V
Measurement Settings: Main Setup -> Measurements -> ST Analysis
For Alarm Mode = Single-ST
For Alarm Mode = Multi-ST
Selecting Leads for ST Analysis
You select which leads to use for ST analysis in the Setup ST Analysis menu.
To select a lead for ST Analysis,
Select Main Setup -> Measurements -> ST Analysis to enter the Setup ST Analysis
Select Setup ST Leads to open the Setup ST Leads menu. All leads currently chosen for ST
monitoring are listed here.
Select the Add key. This opens the Choices pop-up window.
Choose a lead from the list. This closes the Choices window and adds the selected lead to the list of
To disable ST monitoring for a lead,
In the Setup ST Leads menu, choose a lead from the list.
This setting lets you switch ST Analysis On or Off.
MP2 - MP90
M Profile Adult
x +2.0 mm
x -2.0 mm
x +1.0 mm
x -1.0 mm
4 Configuration Settings Appendix
, MCL Settings