4 Configuration Settings Appendix
Profile Settings: Main Setup -> Profiles
Factory default settings for Profiles depend on the monitor model, as well as the monitor's H and A
option. For detailed information on all factory-provided default Profiles, see the section "Configuration
Overview", starting see on page 222.
The monitor does not need a Profile to start monitoring. If, in the case of an error, no configuration is
loaded, or if a loaded configuration is corrupt, the monitor will be operational and use the factory defaults
documented in the configuration tables of this guide. The Paced status will be set to Yes and the
Patient Category will be set to Neo. An appropriate configuration should then be loaded onto the
monitor using the IntelliVue Support Tool.
The default Profile is used after discharging a patient, leaving demonstration, or when the monitor is
switched off for more than 60 seconds (if the global setting Automat. Default is set to Yes).