Telemetry Device Use Models
Absence of successful association
CENTRAL: TELE ONLY INOP
CHECK PAIRING INOP
??? appears as telemetry device label
in assignment window at bedside
A complete list of INOPs can be found in the IntelliVue Cableless Measurement Instructions for Use or
IntelliVue Information Center Instructions for Use.
7 Telemetry Device and CL Pods Use Models
The bedside monitor and the
telemetry device are not
communicating with each other.
Monitoring at the Information Center
When the telemetry device is wirelessly
paired with an X2 patient monitor (no
label) docked with a larger networked
MP series monitor, and the network
connection is lost.
An X2 patient monitor with a
telemetry device label is paired with a
larger networked MP series monitor
and subsequently the telemetry device
is paired with the same monitor. Only
one telemetry device can pair with the
The equipment label is invalid or there
is no label assigned.
• Repeat the device assignment
• Check the telemetry device settings
and ensure SRR is started.
• Restore the monitor's network
• Select the correct device to be
• Contact service personnel.