6 IntelliVue Cableless Measurements
Repeat Time This setting defines the Repeat Time of the automatic measurement. Choices are
from 1 min up to 4 hours.
Label This setting lets you select the SpO
user defaults of the new label.
Signal Quality Set Signal Quality to Off, if you don't want the signal quality indicator to
be displayed next to the SpO
Average The SpO
Average lets you adjust the averaging time between 5, 10, and 20 seconds. It represents the
approximate time period used for the calculation. The exact averaging algorithm depends on the signal
conditions. The longer the averaging time, the longer the time needed until the SpO
physiological event. Fast averaging is useful for situations where an extremely fast measurement is required
or few artifacts are expected. Use slow averaging where you expect the number of artifacts to be relatively
high. Whether Average can be changed in Monitoring mode depends on the setting below.
Average in Mon. This setting determines whether the setting Average can be changed in
Monitoring mode. Set Average in Mon. to Yes to enable the user to change the averaging time in
NBP Alarm Suppr. Set NBP Alarm Suppr. to On to suppress INOPs that would otherwise be
generated when you measure NBP on the same limb as SpO
On, the CL Pod automatically remembers the SpO
INOPs while the cuff is inflated.
Color This setting determines the color of the SpO
numeric on the Screen.
numeric represents an average value calculated from several SpO
measurement label. The following choices are available:
l. Changing the label resets all label dependent SpO
. If NBP Alarm Suppr. is configured to
value measured before cuff inflation and suppresses
measurement displayed at the patient monitor's
Configuring Profiles and Settings
value reflects the