Configuring Profiles and Settings
Modifying a Setting
To change settings in an existing Profile:
Select the Profile (A-D) you want to change and select the Load pop-up key to activate it.
Make the changes to the individual measurements or CL Pod settings.
Select Profile again and select Store to save the settings to this Profile.
To reset the selected Profile to factory defaults, select Reset To Factory.
Changing the CL Pod's Default Profile
Every CL Pod has one default Profile (Profile A). This is marked with a black diamond.The CL Pod loads
the default Profile:
• when you exit Demonstration mode (but not after exiting Configuration or Service mode).
• after discharging a patient.
• after being switched off for more than 60 seconds (only if Automat. Default is set to Yes).
• when selecting New Patient or Free Device from the Patient menu.
Profile A is always the default Profile.
Configuring CL Pod SpO
Main Setup -> SpO
IntelliVue CL Pod
Average in Mon.
NBP Alarm Suppr.
CL Pod SpO
Mode Configure Mode to Manual to allow SpO
request and not continuously; helping to save the CL Pod's battery power. When you change to Auto, a
measurement cycle is automatically started with the set Repeat Time. When date and time are
available, the measurement is synchronized to the clock.
C M Factory Default Setting
x 15 min
6 IntelliVue Cableless Measurements
measurements from the CL Pod to be made on