Select the pop-up key Setup Meas. to open the Setup Measurement Values window, then select
the pop-up key Add.
In the Setup Measurement submenu, select Label and choose the required measurement label from
the pop-up list.
Select the Unit and Color that should be used for the chosen label.
Select Interval to define the time after which a manually entered value becomes invalid (no value
is then displayed). Values can be entered up to two hours after they have been measured or up to the
measurement interval, if this is shorter.
Select Msmnt to define whether the selected measurement will be On or Off. The On/Off state can
be changed by the user in Monitoring mode.
If available, select Format to define an input format. For pressures, for example, you can configure
whether the user should enter all pressure values (Sys&Dia&Mean) or only the Mean pressure. The
Format can be changed by the user in Monitoring mode.
4 Configuration Settings Appendix