Tone Modulation if you set Tone Modulation to Yes, the pitch of the SpO
with the measured signal strength. If you are using two SpO
which sensor will serve as the source for the tone modulation pitch, by selecting the corresponding SpO
label as the System Pulse source. For example, if you measure SpO
source for the tone modulation, in the Setup Pulse menu, set System Pulse to SpO
Tone Modulation Type This setting lets you choose between Standard and Enhanced.
Standard is the regular Nellcor behavior. The difference in frequency per % saturation change is small
which might make it difficult to hear smaller changes. Enhanced results in a larger (and therefore more
obvious) frequency decrease for each drop in SpO
drops below 70%, the absolute frequency may become so low that the perceived volume will be low and it
will be hard to hear a frequency change per % saturation.
Global Speed The Global Speed setting defines the speed of all non-respiratory and non-EEG
waves on the Screen. The Global Speed setting can be overridden by fixed wave channel speeds. See
see "Configuring the Wave Channel Speed" on page 26.
RespiratorySpeed The Respiratory Speed setting defines the speed of all respiratory waves
, Resp. anesthetic agents and O
overridden by fixed wave channel speeds. See see "Configuring the Wave Channel Speed" on page 26.
EEG Speed The EEG Speed setting defines the speed of all EEG waves (including BIS) on the
Screen. The EEG Speed setting can be overridden by fixed wave channel speeds. See see "Configuring
the Wave Channel Speed" on page 26.
Touch Tone Volume The Touch Tone Volume setting defines the volume of the tone you hear
every time you select a field on the monitor screen. You may want to set this to 0 if you want to operate the
monitor in a quiet environment.
MMS Keys To prevent users from using the hardkeys on the Multi-Measurement Module, set
MMS Keys to Disabled. This setting is not available in the MP2 and X2.
ArrhyOff Message If arrhythmia analysis is switched off, the message "Arrhythmia Off" is
permanently displayed beside the first ECG wave on both the bedside monitor and on the Information
Center. If you do not want this message to appear, you must set ArrhyOff Message to Off.
Meas. Selection. The Meas. Selection setting defines the behavior of the monitor if a
measurement label conflict occurs.
• Set this to Indicator to make the measurement selection icon with question marks your only
indicator of the label conflict.
• Set it to Window to pop up the Measurement Selection window indicating the label conflict.
• Set it to Auto Select. to let the monitor resolve the conflict automatically by assigning a new,
generic label to the newly connected measurement device. Be aware that this behavior only applies to
the following plug-in module measurements:
– Invasive Pressure
Auto Select. does not work on any MMS or X2 measurements.
sensors simultaneously, you can determine
level. Due to the larger steps, when the saturation
) on the Screen. The RespiratorySpeed setting can be
4 Configuration Settings Appendix
tone will change
l and SpO
r and want SpO
r to be the