Auto Reports Configuration Implications
Auto Reports print automatically when a specified trigger occurs. There are two types of Auto Reports:
• Scheduled reports can be set up to print at predefined intervals, starting at a predefined time of
day. The start time you set applies for every following day. For example, if you set a start time of 07:00
and a repeat time of six hours, the first report will print at 07:00 every day, the next at 13:00 and so on.
• End Case Reports print when a patient is discharged using the monitor's End Case function.
An Auto Report can include both a Scheduled Report and an End Case report.
Setting up an End Case Report
Select Main Setup -> Reports -> Auto Reports
Select one of the four available Auto Report "slots" (A, B, C, or D)
Select Report, then select the type of report you want to be printed when a patient is discharged
using the End Case function, for example "VitalsReport".
Set End Case Report to On.
Set Scheduled Rep. to Off, if do not want the monitor to trigger a Vital Signs Report at
If needed, set up the VitalsReport or whatever type of report you used for the EndCase Report.
If you want more than one report to be printed when using End Case on a patient, repeat steps 2
through 6, selecting a different Auto Report (A, B, C, or D) and allocating it a different report
Configuring ECG Reports
ECG Reports Configuration
Monitor Setting: Main Setup -> Reports -> ECG Reports
ECG Reports Configuration Implications
Report Lead Layout The layout of the report can be configured to either International or
ECG Gain This lets you set a defined ECG Gain to determine how ECG waves will appear on the ECG
MP2 - MP90
x 10 mm/mV
x 25 mm/sec
4 Configuration Settings Appendix