Fix Pacer Amplit Set this to On to configure the pacer spikes to have a fixed size, for ease of
identification on the monitor Screen. The spikes are then shown in the background as a dotted line. The
length of the dotted line is fixed to the height of the wave channel and is independent of the actual pacer
amplitude. Set it to Off to display the pacer spikes as a solid line in their original height and the pace
pulse markers next to the pacer spikes.
Monitor Setting: Main Setup -> Calculations
Calc. Time Ref.
BSA Form. Adult
BSA Form. Pedi
BSA Form. Neo
Hemo Press Unit
Gas Press Unit
Calculations Configuration Implications
Calc. Time Ref. (relevant for Hemo Calcs only) The calculation time reference determines the
timestamp of all measurement values that are used for a hemodynamic calculation except the C.O. value.
Select C.O. Time if you want to use the values measured at exactly the time of the most recently saved
Select Current Time if you want to use the currently measured values (i.e. the values measured at the
time you entered the Calculations window). The most recent available C.O. value is used for the
calculation. Be aware that if Calc. Time Ref. is set to Current Time, and the calculation is
performed significantly later then the C.O. value has been saved, the difference in timestamps may reduce
the clinical validity of the calculation results.
MP20 - MP90
not applicable, these settings are stored as Global Settings, see "Global Settings" on page 168.
4 Configuration Settings Appendix