gjl 29/01/1999 C:\WINNT\Profiles\dapowell\Desktop\D.0 Books\Books for CD\Source Files\Aq D.0\aqbook.fm
Setting up Physios
A connected ECG is required before you can obtain AQ data. You can display up
to two additional physios in AQ waveform mode and one additional physio
(Pulse) on the Global Function Report. You will find full details of system physio
setup, including cable connection, in your System Basics guide.
After you have connected the proper cables for the physios you want to display,
follow these instructions to setup physio display in AQ:
, and on the left touch panel, touch Physio.
2 Turn the rotary controls to select the desired sources for the physios you want
Set ECG to Normal if you are using the system ECG.
Set ECG to Aux1 if you are using an external ECG source.
Be sure to set a source for any other physios you will want to use in AQ.
These are the physios that are available for all modes, not just AQ. Any physio
turned off here will not be available in any mode.
3 On the right touch panel, touch AQ.