adhesive pads on a yellow plastic liner. Lift the pads out of the cartridge
and unwind the wires from the well. Peel one pad off the liner.
NOTE: Avoid touching the pad's adhesive gel, because this may cause
the Trainer to begin analysis prematurely.
5. Place the pad on the manikin.
Press the pad down firmly. Then
peel and place the other pad.
6. Continue to follow Trainer
instructions for remainder of
7. When the scenario has been
completed, or to stop a scenario
in process, press and hold the On/Off button until the Trainer turns
8. To repeat the same scenario, repack the pads as described below, then
begin again by pulling the green handle or pressing the green On/Off
button. To select a different scenario, press the i-button.
NOTE: If battery power is depleted, the Trainer will say REPLACE
BATTERY when the Trainer is turned on, at the completion of a training
scenario, and when the Trainer is turned off. The Trainer will continue
to operate for a short time. Replace the batteries as soon as possible to
ensure correct Trainer behavior.
STANDARD TRAINING SCENARIOS
The Trainer provides eight training scenarios when using the training pads
cartridges, as described in the following table. When the Trainer directs
you to press the Shock button, the scenario will not advance unless the
button is pressed. When the Trainer detects a non-shockable rhythm, it
instructs you to provide CPR. In the Trainer scenarios, "conversion" means
a change from a shockable to a non-shockable rhythm.