REMstar Auto A-Flex
IMPORTANT! Remove this guide before giving the device to the patient. Only medical professionals should adjust pressure settings.
This guide provides you with instructions on how to access and navigate the provider screens used to modify device settings.
Refer to the User Manual for more information.
Note: The screens shown throughout this guide are examples only. Actual screens may vary slightly.
Accessing the Provider Mode Screens
Accessing provider mode unlocks settings that cannot be modified by the user. To access provider mode:
1. Supply power to the device. First, plug the socket end of the AC power cord into the power supply. Then plug the pronged
end of the AC power cord into an electrical outlet that is not controlled by a wall switch. Finally, plug the power supply
cord's connector into the power inlet on the back of the device.
2. Once the device is powered, the Home screen appears, shown below. Turn the control wheel to toggle between the four
options and highlight "Setup".
Note: "Flex" shown above will display as the current Flex mode.
3. Once "Setup" is highlighted, press and hold both the control wheel and the Ramp button
4. You will hear a quick double beep and the Provider mode screen will appear, shown below. You are now in provider mode.
on the device for at least 5
- 1 -