Select Show ST Limits to expand the list of ST leads and view the currently set alarm limits.
Selecting Hide ST Limits hides the list again.
You can use the pop-up keys that open with the Alarm Limits window to perform common tasks:
– All Al. On/All Al. Off,
– All Lim. Narrow/All Lim. Wide to set narrow or wide alarm AutoLimits for all
– Print Limits/Record Limits to print a list of all current alarm limit settings on a
connected printer or recorder.
These pop-up keys are not available in the window for changing individual alarm limits which you
access by selecting the measurement label in the Alarm Limits window.
Changing Alarm Limits
To change individual measurement alarm limits using the measurement's Setup Menu,
In the measurement's setup menu, select the alarm limit you want to change. This calls up a list of
available values for the alarm limit.
Select a value from the list to adjust the alarm limit.
Alternatively, you can use the keys in the measurement Change Limits window, which you access by
selecting the measurement label in the Alarm Limits window.
Measurement labels, with alarms off symbol where appropriate
Current alarm limits
Graphic view of current yellow and red alarm limits and currently monitored measurement value.
indicates the measurement is switched off