Alternatively, you can use the keys in the measurement Change Limits window, which you access by
selecting the measurement label in the Alarm Limits window.
To change alarm limits,
Enter the Alarm Limits window.
To set the high alarm limit, select the high yellow alarm field to open a pop-up list of high alarm
limits. Select a limit from the list. Repeat to set the low yellow alarm field.
If you set the yellow alarm limit outside the red alarm limit, the monitor will automatically set the red
alarm to the yellow alarm limit.
High red alarm (view only)
High yellow alarm field
Select to open a pop-up list of high alarm limits
Alarms On/Off key - select to toggle between alarms on or off
Preview Alarm AutoLimits for a measurement before applying
Select to apply wide AutoLimits
Select to apply narrow AutoLimits
Low yellow alarm field
Select to open a pop-up list of low alarm limits
Low red alarm (view only)
Graphic view of alarm limits with currently measured value