2 Basic Operation
MP2 Screen Elements
Using the Touchscreen
Touch a screen element to get to the actions linked to that element. For example, touch a measurement
numeric and the setup menu for that measurement opens. Touch a wave to enter the setup menu for
Measurement Setup Menus
Each measurement has a setup menu where you can perform operations or change settings. Typically,
the setup menu window covers the whole screen, with the exception of the INOP and alarm message
fields, which are always displayed at the top. The following picture is for illustration purposes, and may
not exactly represent what you see on the screen. We are using non invasive blood pressure as an
example, but all measurement setup windows are similar and share the same basic layout and
Touch the measurement numeric on the screen to enter the setup menu.
Auto 15 min
Battery status indicator
Touch the numeric to enter the measurement
Touch the wave to enter the measurement
Shows information and messages prompting
you for action.
Gives information about remaining battery
charge, estimated operating time, maintenance
requirements and malfunctions. See the
chapter "Using Batteries" on page 217.
Measurement setup menu
No Central Monit.
NBP meas. + autom. cycle started
Al. from :
Operating and Navigating