For central recording from the bedside, your monitor must be connected via a network to an
Information Center. You can use either the M1116B 2-Channel Recorder or the standalone M3160A
4-Channel Recorder. See the documentation supplied with the Information Center for information on
the 4-Channel Recorder.
Starting and Stopping Recordings
The recordings pop-up keys let you start and stop recordings. Select the Main Setup SmartKey then
select Recordings to call up the line of pop-up keys. Scroll right or left to see any pop-up keys not
Recording Without a Template
To record without a preconfigured template, selecting the waves you require,
Select the Main Setup SmartKey then select Recordings.
Select the pop-up key Select Waves,
Use the pop-up keys to choose up to three measurement waves to be printed on the recording. If
you want fewer than three waves on the recording, select the waves you want then select the
Continue pop-up key.
Select one of the recording speed pop-up keys to set the required recording speed.
Select the Start pop-up key to start the recording.
Select Waves recordings use default values for any recorder settings not defined: runtime is
continuous, overlapping is set to non-overlapping.