15 Assigning Two Devices to One Patient
General Telemetry-related Functions
If later the patient is disconnected from the telemetry device, and reconnected to the monitor again,
any changes in the settings made in the meantime will be passed on to the monitor. In this way,
settings continuity is preserved when the ECG source changes.
Settings synchronization can only take place when there is no patient information mismatch between
the monitor and the Information Center. If a Check ECG Settings or CHECK PAIRING
INOP appears always check that the ECG settings, especially the paced setting, are appropriate for