15 Assigning Two Devices to One Patient
Functions Available For Devices Connected Via
The Telemetry Device window can be opened from the Measurement Selection
window, by selecting the Tele pop-up key, or via Main Setup -> Telemetry Device.
The window shows the equipment label of the assigned telemetry device and, for monitors with SRR
capability, a battery status symbol
If the monitor has a connection to an Information Center, the pop-up keys Unpair to Mon. and
Unpair to Tele let you end the device pairing. After unpairing, the Information Center will
receive data exclusively from the monitor or from the telemetry device.
If the monitor does not have a connection to an Information Center, or there is a pairing mismatch,
the Unassign Tele pop-up key lets you end the telemetry device assignment and close the SRR
Temporarily Stopping the Short Range Radio Connection
To temporarily disable the SRR connection for this device, for example, in case of interference:
Select the Stop SRR pop-up key.
This closes the SRR link and the telemetry device switches over to standard telemetry transmission.
The battery status symbol will no longer be shown and the SRR quality indicator symbol will be
To return to SRR use:
Select the Start SRR pop-up key and press the Check button on the telemetry transceiver.
Setting Up Measurements
The Setup ECG and Setup SpO
The Measmt. Select. pop-up key takes you directly to the Measurement Selection window where
all connected measurement devices are shown.
General Telemetry-related Functions
The following functions are valid for directly and indirectly connected telemetry devices.
Functions Available For Devices Connected Via SRR
and a symbol indicating the signal quality of the
T pop-up keys give you access to the measurement setup menus.