15 Assigning Two Devices to One Patient
WARNING All data presented in the telemetry data window are delayed for several seconds.
If you need realtime data, for example for defibrillation, always use the monitor ECG instead of
telemetry. As long as the ECG is being measured with the telemetry device there will be no ECG signal
available at the ECG Sync Pulse output.
Direct Connection - Automatic Pairing
The following direct connections are possible:
• a TRx+4841A Intellivue Telemetry system transceiver connected via short range radio (SRR) link to
an MP2 with SRR capability.
If a telemetry device is assigned to a monitor via a direct short range radio link, the monitor and the
telemetry device are automatically paired at the Information Center (if configured).
The measured data from the telemetry device — ECG and, if available, SpO
monitor screen and will be sent through the monitor to the Information Center. The ECG waves and
numerics appear in place of the monitor's own ECG, and SpO
measurement. When ECG is measured with the telemetry device there will be no Respiration
• When ECG is being measured with a telemetry device directly connected to the monitor, there will
be no ECG signal available at the ECG analog output or ECG Sync Pulse output and no
synchronization marks on the ECG wave. A No ECG Out message will appear in the ECG wave
• When a telemetry device is connected to the monitor, arrhythmia relearning is initiated, and again
when the telemetry device is disconnected.
** HR 140>120
SOME ECG ALRMS OFF
Auto 10 min
* TELE ALARM
How Can You Combine Devices?
T— will appear on the
T is displayed as an additional