To set the wireless network channel for the MR400 cart
1‐28 Important Information Expression MR400 Instructions for Use
Care should be taken to guard against inadvertent changes to the network channel
setting. Before use, always ensure that all devices are communicating properly. Failure
to do so may cause a lapse in patient monitoring.
An MR400 system is comprised of one MR400 cart, one wECG module, and one
wSpO2 module, and optionally an IP5. In environments where multiple MR400 MRI
patient monitoring systems are being used, you must be aware of each component's
network setting. Operating multiple MR400 systems on the same network or with a
wrong network setting will interfere with communications, and incorrect or corrupted
patient vital signs information will be displayed as a result.
If a patient is currently admitted, a warning dialog box will prompt you before a change to
the monitor network is allowed.
After changing the wireless network channel of the MR400, you must wait a minimum of 5
seconds before removing power from the system; otherwise, the change will be lost.
Press the power switch (see page 2‐7 for the location) then allow
the MR400 to initialize.
The main screen appears.
Select the network icon.
The network menu appears.