Do not use a damaged battery. Periodically check batteries, stop using and replace any
battery that exhibits abnormal heat, odor, color, deformation, or other condition. If a battery
is punctured or if battery liquid leaks onto your skin or clothing, immediately wash the area
and clothing with fresh water. If battery liquid gets into your eyes, do not rub your eyes;
immediately flush your eyes with clean water and consult a physician.
Examining the Shipment
To report shipping damage, or to resolve any issues or concerns with your order, contact
Customer Service. (Save all packing materials and related shipping documents, as these may be
required to process a shipping damage claim with the carrier.)
After removing the contents from the shipping containers, carefully examine all items for signs of
damage that may have occurred during shipment. Also, check all items against the included
packing list and the purchase request.
The contents of the crate should include:
A separate container may include additional items:
Expression MR400 Instructions for Use Important Information 1‐13
If the battery contacts become dirty, wipe them clean with a dry cloth before use. Do not
immerse in a battery in water or other liquids.
Store batteries in a dry place, between 0 to 40°C (32 to 104°F) . Do not expose a battery to
temperatures above 60°C (140°F).
Do not short the external battery contacts. Keep metal objects away from the battery
Store each battery in a manner that prevents shorting with the container or another cell/
Only use the Philips specified charger.
Expression MR400 MRI Patient Monitoring System
Two main batteries
Instructions for Use (IFU) manual
Quick Reference Guide (included in English‐localized shipments only)
Wireless ECG patient module (Gen 3)
Wireless SpO2 patient module (Gen 3)