Sets the mode used to take NIBP readings.
A manual reading will not restart this cycle time.
The following options are available:
To set the mode for NIBP readings
12‐16 Monitoring NIBP Expression MR400 Instructions for Use
The Interval menu appears. The current setting is highlighted.
Select the desired minute(s) for the interval:
The setting is changed.
Off takes readings manually (Manual mode), where readings are initiated by pressing the
NIBP Start/Stop key; see Making Manual Measurements on page 12‐7 for details. (Default)
On takes readings automatically, where when selected (or since leaving suspend mode),
the first reading must be initiated by pressing the NIBP Start/Stop key and then all
subsequent readings will be taken at the selected interval; see Making Automatic
Measurements on page 12‐6 for other setup details.
Select the NIBP VS box.
The NIBP menu appears. Current settings are displayed.
Locate Auto Mode and select the desired setting:
The setting is entered.