Auto Mode setting
NIBP VS box label
Changing the NIBP Format
To control the format of the NIBP data
Changing the Unit of Measure
When using an IP5 and Pressure Units is changed, the displayed formatting of the value and
placement of the decimal point is changed immediately. However, it can take up to 2 seconds for
the measurement numeric values to reflect the new unit of measure. Do not print or perform data
captures during this period.
Expression MR400 Instructions for Use Monitoring NIBP 12‐11
Indicates Manual when in manual mode; or, Next when in automatic mode along
with the time until the next NIBP measurement, displayed in the following
<hh> = Two-digit hours field
<mm> = Two-digit minutes field
<ss> = Two-digit seconds field
Indicates the NIBP vital sign parameter, and accesses the NIBP menu
Select the NIBP VS box.
The NIBP1 menu appears. Current settings are displayed.
The Format menu appears. The current setting is highlighted.
Select the desired format:
The format is changed.