Stopping an NIBP Measurement
Press the NIBP Start/Stop key to stop a reading cycle.
Suspend Mode during NIBP Measurements
When the Suspend key is pressed, NIBP functions will be affected as follows:
NIBP VS Box
The NIBP measurements are displayed as numeric information in the NIBP VS box. Other data,
including NIBP‐related alarm information, are also provided in this area of the screen, as detailed
Depending upon the selected Format (see page 12-17) of the data, the elements contained in the
NIBP VS box are displayed in the Systolic/Diastolic format or in the Mean format.
12‐8 Monitoring NIBP Expression MR400 Instructions for Use
Any reading in progress will be stopped.
The system will pump to the initial inflation pressure for the selected patient type.
Manual readings can be taken.
Auto readings cannot be taken.
The "NEXT" timer will not run.
When exiting Suspend Mode, any manual reading will not be stopped.