Choosing the Measurement Mode
NIBP measurements can be taken automatically or manually, using the mode that best suits the
needs of your patient. The following are frequently used functions related to NIBP; also refer to
page 12‐14 for other NIBP functions.
Making Automatic Measurements
You can automatically measure a patient's blood pressure at predefined intervals, which are
measured from the start of one NIBP measurement to the start of the next.
To turn on automatic operation
12‐6 Monitoring NIBP Expression MR400 Instructions for Use
Attach a NIBP cuff appropriate for the patient type and size to the
Position the cuff on the patient; see page 12‐5.
Ensure that the cuff and interconnect hoses are not kinked.
Select the Patient Type.
See Selecting the Patient Type on page 3‐11.
Press the NIBP Interval key.
The Interval menu appears. The current setting is highlighted.
Select the desired minute(s) for the interval:
The selection is entered.