AGENT and GAS VS Boxes
AGENT option measurements are displayed in several areas:
9‐14 Monitoring Agents and Gases (AGENT Option)
With the new water trap positioned so that the sample port is facing
toward you, align the trap's release tabs to the release tab slots in the
water trap receptacle. Press the water trap into the water trap
receptacle until audible "clicks" are heard as both of the release tabs
lock into the release tab slots.
Attach the sampling line to the sample port.
Primary and secondary agent gas measurements are displayed as numeric information in
the AGENT VS box.
N2O, O2, and MAC measurements are displayed as numeric information in the GAS VS box.
CO2 measurements are displayed as a waveform in the VS trace area of the screen and as
numeric information in the CO2 VS box—where depending upon the RESP > Source
setting, CO2‐derived respiration information may appear in the RESP VS box.
Release tab slots
Water trap receptacle
Expression MR400 Instructions for Use