Disposable attachments are designed for single patient use and must be disposed after use.
They must not be cleaned and reused. Follow your hospital's guidelines for appropriate
disposal. Reuse of single-use devices can result in spread of patient infection, degradation of
monitoring performance, or inaccurate measurements.
Use only approved pressure transducers and cables, as listed in Accessories in chapter 1, and
follow the Instructions for Use that are supplied with the pressure transducer to set up and use the
transducer monitoring kit.
The procedure below details the connection of a single transducer. To monitor two IBP channels,
repeat the procedure to connect an additional transducer to the unused IBP port on the MR400.
Adult and Pediatric Patients: Expression MR IBP DPT Kit, A/P (REF 989803194631)
I. Connecting the Reusable Cable to the MR400
Connect the reusable cable, REF 989803194601 (or equivalent), to the P1 or the P2 port on the
patient connection panel.
8‐4 Monitoring Invasive Blood Pressure
(REF 989803194601, or equivalent)
Expression MR400 Instructions for Use