Press the Setup key and then the
Monitor key. On the Monitor Setup
menu, select Service(Bio‐Med). On the
Service(Bio‐Med) menu, select NIBP
The following menu items are
If an error is reported while the Leak Test is in progress, the test will be canceled and user will be
prompted with the message: NIBP Bench Error. Press Start to try again.
To perform these tests
See reference the service manual for details.
Adjusts the brightness of the touch screen.
To open the Backlight Brightness menu
3‐34 Getting Started Expression MR400 Instructions for Use
High Range Check
Calibrate performs calibration of the NIBP system. (Password required)
Static Test performs a static pressure test of the NIBP system. (Password required)
Leak Test performs a pressure leak test of the NIBP system (see note below).
Stress Test performs a pressure stress test of the NIBP system. (Password required)
High Range Check performs a high pressure test of the NIBP system. (Password required)