Once the source is located, attempt to attenuate the interference by distancing
the transceiver from the source as much as possible. If assistance is needed,
contact your local service representative.
Artifact on ECG and other physiological waveforms caused by electromag-
netic interference should be evaluated by a physician or physician authorized
personnel to determine if it will negatively impact patient diagnosis or treat-
Safety Standards & Specifications
Is the interference intermittent or constant?
Does the interference occur only in certain locations?
Does the interference occur only when in close proximity to certain
medical electrical equipment?
Do parameter measurement values change dramatically when the AC line
cord is unplugged?
RF Output Power
RF Output Power control
Bands 1395-1400 MHz and 1427-1432 MHz
Channel Spacing: 1.6 MHz
8 dBM +2/-3 dB (3.2 mW to 10 mW), into
Antenna load @ nominal battery voltage
• Transceiver chain gain shall be calibrated
at time of manufacture (env temp 25+/- 5
• Over power algorithm deployed to prevent
excess power and also includes