Observe the screen for cable and lead errors as follows.
When all the leads are properly connected, the ECG waveform appears.
Use the arrow buttons to change the gains settings of the display, as shown in the
N O T E
The recording for all resolutions is always made at the 1X gain setting.
Gain on screen
Use the arrow buttons to select the
N O T E
If you do not start the recording, the DigiTrak XT automatically starts recording after 30 minutes,
provided there are no lead errors.
During a Recording
During a recording, the DigiTrak XT displays the date, current time, and the time remaining
for the recording as well as any lead or cable error, if present.
± 5 peak-to-peak
± 2.5 peak-to-peak
± 1.25 peak-to-peak
, if applicable.
tab and press
DigiTrak XT Recorder Instructions for Use
to start the recording.