If the MX40 experiences loss of network connectivity, technical alerts at the
Information Center ("No Signal") and at the MX40 ("No Central Monitor")
will occur. The MX40 will also automatically revert to local monitor mode
which activates display of patient data on the MX40 – however, when in
this state, battery life will be shortened.
Safety Standards & Specifications