An MX40 operating with Release L or M of the Information Center does not
have physiological alarm capability locally at the device (networked or
non-networked). A No Alarm Display message is present along with the
Alarms Paused icon.
When operating with Information Center Release L or M, the alarm
pause time of the MX40 is not configurable. The alarm pause time for
the MX40 is always two minutes.
When operating with Information Center Release L or M, if alarms are
paused at the Information Center, the "Alarms Paused" message is only
displayed at the Information Center.
Technical alarms (INOPs) are communicated and can be silenced using the
Alarm Silence button.
Technical alarms can be reviewed using the Alarms SmartKey.
Note — Not all rechargeable battery technical alarms are communicated via
the Alert Data Integration paging system. The following alarms are not
TELE CHECK BATT
TELE SERVICE BATT
TELE BATTERY TEMP
However, these technical alarms are still transmitted to the Information
Operating with Information Center Release L or M