10. Monitoring with other
This section provides information about the use of the MX40 when it is
assigned to other monitoring devices. The MX40 can be assigned to
IntelliVue Patient Monitors or IntelliVue Cableless Measurements for SpO
and NBP. The connection to these other devices is done by pairing
networked devices or using the integrated short-range radio of the MX40.
For additional information on IntelliVue Patient Monitor or IntelliVue
Cableless Measurements operation, consult the Instructions for Use that
accompanied the device.
Assignment of the MX40 to IntelliVue Patient Monitors is only supported
when the patient monitor (MP5,MP5T,MP5SC, MP2 or X2 only) is equipped
with a short-range radio. Monitors that have this equipment will display
the short-range radio symbol
When the MX40 is connected to a patient monitor using the MX40 /
IntelliVue Patient Monitor Adapter Cable, (p/n 989803172211), the SpO
sensor must be directly connected to the patient monitor to monitor SpO
Note — Assignment of the MX40 to IntelliVue Patient Monitors is not
available with patient monitors connected to the M3140 Information Center.
Note — The MP5T and MP5SC are non-networked devices as they does not
support a connection to the Information Center.
on the label.
Monitoring with other Assigned Devices