4 Managing Patients
printed when you select
up end case reports.
New Patient Check
The monitor can be configured to ask you in certain situations:
after a specified power-off period
after a specified standby period
when no basic vitals (HR, RR, Pulse, SpO
whether a new patient is now being monitored. The pop-up window is entitled
. The monitor offers a
patient and a
The time periods for the three conditions can be configured independently.
To save you from having to enter the same patient data multiple times and enable patient transfer
without loss of data, information can be shared between Multi-Measurement Modules (MMS), patient
monitors, and Information Centers.
patient demographic information is shared between connected MMSs, patient monitors, and
measurement settings and calibration data can be uploaded from an MMS to a patient
monitor, if configured
trend information can be uploaded from an MMS to a patient monitor, if configured.
Different sets of patient and measurement-related data are stored in the monitor and the Multi-
Measurement Module. Understanding this will help you to understand what happens to patient data
when you transfer patients.
Patient demographics (name,
DOB, patient IDs)
Monitor settings (alarm
pause time, alarm volume)
- to discharge the patient without printing any reports.
key to discharge the previous patient and begin monitoring a new
key to continue monitoring with the current patient data and settings.
Stored in Monitor
. See the section on AutoReports for information on setting
, NBP) have been measured for a specified period
Is This A New
Stored in MMS and