All the currently available measurement devices, for example MMSs, MMS extensions, measurement
modules, Gas analyzers, devices connected to a host monitor and manually entered measurements
), are depicted in the
Any measurement labels causing a label conflict are shown in red. If a measurement device is
connected but currently unavailable, for example, because it was deactivated due to a label conflict, the
device is shown "grayed-out".
When an X2 or MP5 is connected to a host monitor, the measurement selection window looks like
X2 connected to host monitor
When an X2 or MP5 with an active measurement, say SpO
same measurement already active, the SpO
INOP is displayed. The measurement can only be reactivated if the X2 or MP5
is disconnected from the host monitor. The label conflict can be resolved on the host monitor like any
other label conflict.
To resolve a label conflict:
Select the measurement selection key or select
Select the device whose label you want to correct.
Use the measurement selection pop-up keys to resolve the conflict. Select either:
: to assign a different label to the conflicting label
MP5 connected to host monitor
measurement on the X2 or MP5 is deactivated and the
De-activated devices are grayed-out (here
the Temp measurement on the left)
, is connected to a host monitor with the
to display the