Choose a label from the list.
The monitor automatically applies the scale, color, etc. settings stored in the Profile for the label you
select. You can change scale settings in Monitoring Mode, but color can only be changed in the
monitor's Configuration Mode.
Any labels already being used in the monitor are shown "grayed-out" in the list and cannot be selected.
About Label Sets
Your monitor may be configured to use a
labels for Pressure and Temp. See the sections in this Instructions for Use on Pressure and Temp for
If you connect an MMS from a monitor using a
monitor. This may cause a label conflict with other monitored measurements.
Be aware that connecting a monitor using the
software revisions may affect the availability of measurement information from the additional labels on
the Information Center. See the Information Center documentation and your monitor's Configuration
Guide for information on label set compatibility.
Resolving Label Conflicts
Each label must be unique, that is, it can only be assigned once. You cannot monitor two pressures
labeled "ICP" at the same time. If you need to use two identical pressures, you must assign different
labels to them, for example, P and ICP.
Measurement labels are stored in the measurement device (module or MMS). If you try to use two
measurement devices that have identical labels, this causes a label conflict in the monitor.
Measurement selection key with question marks indicating a label conflict.
Depending on your configuration, the monitor will either
resolve the conflict automatically, by assigning a new, generic label to the most recently connected
conflicting label (e.g. a second FAP label could be changed to ABP)
take no action, you must enter the
label set or an M3/M4 monitor, any additional labels switch to labels available in the target
label set. The
label set to an IntelliVue monitor using a
label set to an Information Center with certain
window automatically for you to resolve the conflict
window and resolve the conflict.
label set provides extra