This section lists accessories for use with Philips SpO
technologies, refer to the instructions for use provided with these devices.
Some Nellcor sensors contain natural rubber latex which may cause allergic reactions. See the
Instructions for Use supplied with the sensors for more information. M1901B, M1902B, M1903B and
M1904B disposable sensors are not available in USA from Philips. Purchase Nellcor sensors and
adapter cables directly from Tyco Healthcare.
Do not use more than one extension cable with any sensors or adapter cables. Do not use an extension
Masimo adapter cables,
Philips reusable sensors or adapter cables with part numbers ending in -L (indicates "Long"
version) or with a cable longer than 2 m.
All listed sensors operate without risk of exceeding 41°C on the skin, if the initial skin temperature
does not exceed 35°C.
Make sure that you use only the accessories that are specified for use with this device, otherwise patient
injury can result.
Always use the MAX-FAST forehead sensor with the foam headband provided by Nellcor.
Option A01 is the Philips FAST-SpO
Philips reusable sensors.
Adult sensor (2 m cable), for patients over 50
kg. Any finger, except thumb.
M1191A/B with longer cable (3 m)
Small adult, pediatric sensor (1.5 m cable) for
patients between 15 kg and 50 kg. Any finger
Neonatal sensor (1.5 m cable) for patients
between 1kg and 4 kg. Hand or foot.
Ear sensor (1.5 m cable) for patients more than
Infant sensor (1.5 m cable) for patients
between 4kg and 15 kg. Any finger except
Adult clip sensor (3 m cable) for patients over
40 kg. Any finger except thumb.
Option A02 may not be available in all countries. Some sensors may not be available in all
technology. For accessory lists for other SpO
version; Option A02 is the Nellcor OxiMax-compatible
No adapter cable