All ECG Waves Recordings
All ECG Waves
each ECG lead. All available leads are recorded sequentially in the standardized lead order. The
recording is realtime, that is, the information recorded is not simultaneous.
Creating and Changing Recordings Templates
To save you defining recording settings each time you start a recording, you can create templates for
commonly-used types of recordings in the
delayed recording, one alarm recording, three realtime recordings (two for MP20/30), and one high
Changing recordings templates changes the settings that will be used each time a recording of this
name is triggered.
Select the name of the template you want to create or change. Each recording name is linked to a
recording type, delayed, alarm, realtime, and hi-res trends. Recording names can be changed in the
monitor's Configuration Mode.
Design the template by selecting each menu item and entering the information for the template.
assigned to a recording channel in a particular template is not available when a recording is
triggered, the channel is left blank on the recording strip. The pop-up list of available (currently
monitored) waves differs according to the recording type:
Realtime, delayed and alarm recordings: the list shows all the currently available waves.
High-resolution recordings: the list shows all the available high-resolution waves.
In addition to the currently available waves, you can choose from several other settings which
make an automatic allocation when the recording starts:
For high-resolution recordings only the
before the recording is started. This interval is called the "Delay Time" and can be set as specified
in "Overview of Recording Types" on page 317. This setting can only be changed in Configuration
changed in Configuration Mode. Continuous recordings run indefinitely.
recording shows a 4 second recording of each lead with a calibration bar preceding
All ECG Waves
permanent key to call up the
to enter the
: choose which recorder the recording will print to (
(for realtime reports in alarm recording only)).
, etc.: choose which waveform to record in each channel. If the wave
will always record the measurement in alarm in the chosen recorder channel
will always record the current primary lead in the chosen recorder channel
will always record the current secondary lead in the chosen recorder channel
will always record the currently selected anesthetic agent.
: define whether the recorded waveforms will be printed overlapping or beside each other.
: choose the recording print speed.
: Delayed recordings start documenting on the recorder strip from a pre-set time
: see how long this type of recording is configured to run. This setting can only be
menu. You can create templates for one
recordings do not need to be configured, they always use the
setting is available.