The M1116B plug-in recorder records numerics for all active measurements and up to three
waveforms. You can use it for local recording mounted in the monitor's FMS.
For central recording from the bedside, your monitor must be connected via a network to an
Information Center. You can use either the M1116B 2-Channel Recorder or the standalone M3160A
4-Channel Recorder. Recordings made on the M3160A may look slightly different to those described
here. See the documentation supplied with the Information Center for information on the 4-Channel
Starting and Stopping Recordings
The recordings pop-up keys let you start and stop recordings. Select the
call up the line of pop-up keys. Scroll right or left to see any pop-up keys not displayed.
M1116B Plug-in Recorder
Continue LED - flashes if a continuous recording is
RUN/CONT key - starts a delayed recording or
makes the current recording continuous
STOP key - stops the current recording