19 Assigning a Telemetry Device and a Monitor to One Patient
Enter here the equipment label of the telemetry device to be paired.
Pairing at the monitor is only possible when the monitor already has a connection to the Information
Center and the Information Center software version allows pairing at the monitor.
Telemetry Data Window
The measurements from the telemetry device are shown in the telemetry data window on the monitor.
A current alarm generated by the telemetry device.
Telemetry data window
All data presented in the telemetry data window are delayed for several seconds.
If you need realtime data, for example for defibrillation, always use the monitor ECG instead of
telemetry. As long as the ECG is being measured with the telemetry device there will be no ECG signal
available at the ECG analog output.
Direct Connection - Automatic Pairing
The following direct connections are possible:
an MP5 or X2, declared as a telemetry device, and connected with an MSL cable to a monitor.
When ECG is being measured with a telemetry device directly connected to the monitor, there will
be no ECG signal available at the ECG analog output or ECG Sync Pulse output and no
synchronization marks on the ECG wave. A
menu will appear with only one entry
- indication that data in the window is delayed
No ECG Out
message will appear in the ECG wave