17 Monitoring EEG
Select the required scale from this list.
Scaling information is displayed with each EEG wave.
are shown with the EEG wave.
bar beside the EEG wave.
Changing Filter Frequencies
The low and high pass filters screen out undesirable interference from the raw EEG wave display. The
current EEG filter frequency settings are shown in the header of the CSA. Changing filter settings
affects the EEG wave and all the EEG numerics. Whenever the filter setting is changed, a filter change
marker appears next to the spectral lines.
To change the filter settings:
Select the required frequency from this list.
Changing the Speed of the EEG Wave
The EEG measurement has its own speed control and is not affected by the wave speed settings of the
defines the speed at which the wave is drawn across the screen in millimeters per second (mm/s).
The content of EEG Reports is always the same and does not need to be configured.
To print an EEG Report, in the
Alternatively, you can select the CSA and use the
To modify the buffer and trendline settings on the CSA Report, in the
trendlines for the SEF numeric on and the buffer time from Buffer C.
CSA On EEG Rep.
is set to
is set to
in Configuration Mode, the current wave scale is indicated by a size
. If you do not change these settings, the monitor will use the default settings with the
This menu entry lets you
choose one of the three pre-configured buffer times
switch the trendline of the specific numeric on or off
to call up a list of wave scales.
in Configuration Mode, gridlines and the current wave scale values
. Choose the required speed from the pop-up list. This
to call up a list of available frequencies.
pop-up key to start the report.