The CSA contains the following information
Changing CSA Settings
When CSAs are shown embedded on the Main Screen, select the CSA to show the pop-up keys. When
the CSA is shown in a window over the current screen, the pop-up keys are always displayed.
Setup CSA Window
Changing EEG Settings
Be aware that any changes made to EEG settings apply to both EEG channels.
Switching EEG Numerics On and Off
Each EEG numeric can be individually switched on or off in the
Changing the Scale of the EEG Waves for Display
This only changes the visual appearance of the wave. It does not affect the signal analyzed by the
monitor or printed in reports or recordings.
Buffer label and time
Current SEF Threshold
The energy at each frequency is computed and displayed as a spectral line
EEG values are sampled at configured time intervals and displayed as color-coded trendlines.
Trendlines are available for the three frequency numerics (SEF, PPF, MDF)
This menu entry lets you
choose one of the three pre-configured buffers. A buffer defines the
interval between the spectral lines displayed in the CSA.
switch the trendline of the specific numeric on or off
menu, select the numeric label.
to switch between the settings.
for example, Fp1-T3, Fp2-T4
CSA1 or CSA2 according to EEG channel
for example, Montage A
for example, 1-30 Hz
the buffer and interval between spectral lines on the CSA
can only be changed in Configuration Mode
Filter change marker
Montage change marker
17 Monitoring EEG