5 ECG, Arrhythmia, ST and QT Monitoring
Ectopic Status Message Explanation
(No message displayed)
Pacer Not Capt
Pacer Nt Pacing
During a learning phase:
Alarm timeout periods are cleared
Stored arrhythmia templates are cleared
Asystole, Vfib, and HR alarms (when there are enough beats to compute the HR) are active. No
other alarms are active.
Initiating Arrhythmia Relearning Manually
To initiate relearning manually, in the
While the monitor is learning, the delayed arrhythmia wave displays the beat label
rhythm status message
Next, the monitor determines the dominant rhythm. The beats are labeled
status message changes to
After relearning is complete, you should check the delayed arrhythmia wave to ensure that the
algorithm is labeling the beats correctly.
If beats are still not classified correctly, check that the ECG is optimized for arrhythmia monitoring.
You may need to select a different lead or change the electrodes or electrode positions if there is
excessive noise, unstable voltage, low amplitude, or large P- or T-waves.
No ectopic activity within the last minute
More than 2 consecutive PVCs within the last minute
Pair PVCs within the last minute
Pause with pace pulse (paced patient only) within the last
Pause without pace pulse (paced patient only) within the last
No beat detected for 1.75 x average R-R interval for HR <120,
No beat for 1 second with HR >120 (non-paced patient only),
No beat detected for more than the set pause threshold.
R-ON-T detected within the last minute
Multiform PVCs detected within the last minute
SVPB count within last minute is greater than 5
1-5 SVPBs in the last minute with a sinus rhythm and no Vs
SV count within last minute and rhythm status is paced
Paced beat count within last minute and rhythm status is not
B or E
, and the rhythm