5 ECG, Arrhythmia, ST and QT Monitoring
Setting Up the 12-Lead ECG
changes are needed. All changes apply to both the display and the printed 12-Lead report.
Rhythm Lead 1
Rhythm Lead 2
Rhythm Lead 3
Storing and Sending the 12-Lead ECG
You can enter an order number that will then be associated with the 12-Lead ECG. To enter an order
number, select the
the pop-up keys.
To store the previewed data on the monitor and send it to a connected Information Center for storage
and analysis, select the
Center, use the
the 12-Lead ECG data will not be uploaded from the MMS to another monitor after transport.
If the connection to the Information Center is not active while storing, the 12-Lead ECG will be
automatically sent as soon as the connection is established or restored. If a previously stored 12-Lead
ECG is waiting to be sent to the Information Center, it will be overwritten when a new 12-Lead ECG
is stored and can then not be sent to the Information Center.
Printing the 12-Lead ECG
Print a dedicated 12-Lead ECG report by selecting the
Setup 12 Lead
window, select the
Applies a filter to the waveforms.
Choices are 0.05-150Hz D, 0.5-150Hz eM, 0.5-40Hz M (or 0.5-55Hz M for
pediatric and neonatal patient categories), 0.5-20Hz F. The selected filter will be
displayed in the title bar of the pop-up keys.
Defines the gain used for the waveforms.
Choices are 2.5 mm/mV, 5 mm/mV, 10 mm/mV and 20 mm/mV.
Defines the gain for the chest leads relative to the standard gain setting.
Choices are Full and Half.
Defines the waveform speed.
Choices are 25 mm/sec and 50 mm/sec.
Defines the page layout.
Choices are 12X1, 6X2, 3X4, 3X4 1R and 3X4 3R. 1R and 3R refer to 1 or 3 rhythm
Defines whether all leads show simultaneous or sequential intervals of time, when displaying
or printing 2 or more columns.
Choices are Sequential and Simultaneous.
Defines which lead will be used as rhythm lead 1, 2 or 3.
Choices are Primary, Secondary, I, II, III, aVR, aVL, aVF, V1, V2, V3, V4, V5, V6, V,
This setting is only relevant when 3X4 1R or 3X4 3R is selected as Format.
Defines the sequence in which the leads are presented.
Choices are Cabrera and Internat.
pop-up key. The order number will be displayed in the title bar of
Store & Send
pop-up key. If the monitor is connected to an Information
Store & Send
function before patient transport to store the 12-Lead ECG centrally, as
pop-up key to choose settings for filter, gain, etc., if